Home » Categories » Multiple Categories 
Do ProSoft AFC flow computer solutions support energy calculations according to the ISO6976 standards? 
Article Number: 354  Rating: 5/5 from 1 votes  Last Updated: Wed, Dec 31, 2014 at 4:25 PM 
ProSoft AFC flow computer modules implement the Bernoulli equation for measurement of fluid flow through concentric squareedged orifices. The Coefficient of Discharge in this equation is calculated by an appropriate Standard, which may, by option selection, be chosen as either AGA 3 (1992) or ISO5167 (2003). For energy measurement (heating value, calorific content), the AFC currently uses the equations and values described and specified in Appendix C.4 of AGA 8 (1992), calculating either gross (superior) or net (inferior) ideal energy content (gross/net selected by option) corrected to the user's chosen reference temperature. As for how these standards compare with the specifications of the ISO6976 Standard, please consider the following: 1. Energy (heating value, calorific content): As stated before, the AFC Flow Computers use the methods and values given in AGA 8 (1992) Appendix C.4 to calculate molar heating value. This calculation is fundamentally identical to that of ISO6976, with these caveats: (a) There is some minor variation between the two Standards in some of the numerical values, specifically the purecomponent enthalpies at 25°C. Such numerical properties of pure components must be determined by careful measurement and are expected to be revised from time to time as newer experimental results are accepted as Standard. (b) ISO recommends selecting the purecomponent enthalpies from the tabulated column that corresponds to the user's selected reference temperature, but states that these values were calculated from the 25°C values using one or another method (references given), often involving a polynomial expression of Cp (specific heat at constant pressure) as a function of temperature. ISO's reason for giving tabulated values instead of computation procedure is to simplify the calculations that the user must perform and avoid the need for machine computation (programmed computer). On the other hand, AGA 8 lists the coefficients of a seconddegree (order 2, quadratic) polynomial for Cp over temperature, and the ProSoft AFC Flow Computers implement the programmedcomputer method We do not yet know which, if any, of the ISO calculation references the values in AGA 8 are derived from, but ISO states that there is no significant difference among any of their several referenced methods over the temperature range 0°C to 25°C, which is the range of reference temperature permitted by the ProSoft AFC Flow Computers. The advantage of AGA 8's method is that the user can choose any reference temperature between 0°C and 25°C instead of being limited to a choice among four specific values (though these four values cover the vast majority of all choices for metric reference temperature, and an appendix gives the enthalpies for the fifth temperature of 60°F.) (c) Besides heating value on a molar basis, the AFC also calculates it on both a mass and a volume basis. All three values are characteristic of the fluid, depending only on component analysis and reference temperature (volume depends also on reference pressure); calculations thereof are identical to those specified by ISO (other than units conversion), except for the realgas density used for the volume calculation (see 2 next). Energy accumulations and flow rates are calculated by multiplying the appropriate characteristic value with the increment or flow rate of the primary input (mass or volume) scaled as necessary for units of representation. 2. Density and Compressibility (compression factor): These values are calculated according to AGA 8 (1992) Detail Characterization Method, and not according to the much simpler method of the ISO Standard (which seems to be very similar to that of pre1992 AGA 8). ISO's rationale for using the older method is to "enable tractable calculations to be made without necessitating machine computation" One would expect therefore that greater accuracy can be obtained by using AGA 8, and this expectation is supported by the fact that the compressibilities tabulated in ISO6976 are to only 4 digits for the major components of natural gas and to 3 digits for the less common components; the "summation factors" used in the ISO calculation are derived from those values, so the uncertainty in the resulting compressibility will be relatively high regardless of accuracy. Furthermore, ISO lists values only for three common metric reference temperatures (excluding even 60°F) while AGA 8 permits computation for any reasonable reference temperature, as described in 1(b) above. The only clear advantage of ISO6976 over AGA 8 is the greatly expanded list of pure components. 3. Wobbe index: This is not calculated by ProSoft AFC flow computers. 
Attachments (0)
There are no attachments for this article.

Comments
There are no comments for this article. Be the first to post a comment.

Is it possible to configure an AFC flow computer solutions to use the ISO 6976 computation standard?
Viewed 1 times since Thu, Apr 24, 2008
Do the serial ports on my MVI56 module provide any electrical isolation?
Viewed 1 times since Tue, Nov 28, 2006
Using AFC modules with corrected (net) flow as in input
Viewed 1 times since Fri, Sep 19, 2014
How do I configure a vortex, ultrasonic or coriolis type meter in my AFC module?
Viewed 1 times since Tue, Nov 28, 2006
For what purposes would I use the Multiple Meter Streams options on my MVIxxAFC Flow Computer?
Viewed 1 times since Wed, Mar 19, 2008
How do I include or calculate a natural gas specific gravity value?
Viewed 1 times since Mon, Dec 14, 2009
Do ProSoft AFC Flow Computers support annubartype flow sensors?
Viewed 1 times since Tue, Jul 8, 2008
Downloading a new .AFC project file to an AFC flow computer module.
Viewed 1 times since Tue, Nov 28, 2006
The Wallclock in my ProSoft MVIxxAFC module seems to be showing Greenwich Mean Time (GMT or Universal Time) instead of Local Time. How do I get it to use Local Time?
Viewed 1 times since Wed, Dec 3, 2008
Can an MVI71AFC or a 2100AGA module to work in a remote rack?
Viewed 1 times since Tue, Nov 28, 2006
