For Vortex, Ultrasonic or Coriolis meters, the selection depends on the output generated by
If the meter provides a pulse train representing the volume increment, the AFC meter should be
configured as Linear with Primary Input selected as Pulse Count.
If the meter provides the instantaneous flow rate, then the AFC meter should be configured as
Differential with Primary Input selected as Flow Rate.
MVI46-AFC, MVI56-AFC, MVI69-AFC, MVI71-AFC, PTQ-AFC, flow